[Date Prev][Date Next][Thread Prev][Thread Next][Date Index][Thread Index]
Re: Anyone ever heard of Hanna Meters?
I had a Hanna pHep 3 meter for 2 years and it worked
flawlessly. Joe is right you need to recalibrate the
meters about once every month. Additionally, I used
to soak mine in the pH 4 solution for 10 min prior to
calibration to remove organic compounds from the tank
that might buildup on the glass electrode. Mine
worked for 2 years and then the electrode would no
longer calibrate properly. Since the electrode can
not be removed from the unit, the pH meter was
finished. Fortunately, Hanna's new pHep 4 & 5 have
replaceable electrodes. I plan on getting one of
<"Hanna and Oakton produce a variety of industrial
I have a Hanna HI981411 pH controller (+/- 0.1 pH).
This is a panel mount
controller and costs about $100. I bought it through
Systems. I also have the Hanna Mini pH Checker. You
can get this fom
Aquatic Eco-Systems for $18. Aquatic Eco-Systems has
a good selection of
meters/controllers in their catalog if you want to see
between $18 and $600.
All pH meters/controllers are as good as their
calibration. Every meter
I've tried tends to drift to reading high, that is it
will read a pH higher
than the actual pH (for example reads 6.8 when pH is
6.4). This gave me a
fatal CO2 overdose once. I've found that if I
calibrate and clean the
elctrode in the sump monthly, I'm ok. I check the
controller versus a pH
checker weekly as well.">
Do You Yahoo!?
Send FREE Valentine eCards with Yahoo! Greetings!